Time-Domain Reflectometer Campbell Scientific TDR200
The TDR200 Time-Domain Reflectometer is the core of the Campbell Scientific time-domain reflectometry (TDR) system. TDR systems accurately determine soil volumetric water content, soil bulk electrical conductivity, rock mass deformation, or a user-specific time-domain measurement. One Campbell Scientific data logger can control multiple TDR200 reflectometers.
Benefits and Features
- Low power (half the power of the TDR100)
- High sensitivity
- High resolution
- Low noise
- Advanced waveform filtering
- Advanced waveform analysis algorithm
- Backward compatible with TDR100 systems (CRBasic data loggers only)
- 60 Hz frequency rejection
The TDR200 generates a short rise time electromagnetic pulse that is applied to a coaxial system that includes a TDR probe for soil water measurements. Then the reflectometer samples and digitizes the resulting reflection waveform for analysis or storage.
The elapsed travel time and pulse reflection amplitude contain information used by the on-board processor to quickly and accurately determine soil volumetric water content, soil bulk electrical conductivity, rock mass deformation, or a user-specific time-domain measurement.
The data logger collects a 250-point waveform and analyzes it in approximately two seconds. Each waveform can have up to 10,112 data points for monitoring long cable lengths used in rock mass deformation or slope stability. Advanced noise filtering and averaging make accurate measurements possible in noisy environments.
A Complete System
A complete TDR200-based system includes the TDR200, SDM8X50 multiplexers, data logger, power supply, enclosures, and probes. PC-TDR version 3 software supports TDR200 and sensor setup, troubleshooting, and program generation. This software is available, at no charge, from our website.
The SDM8X50 Multiplexer brochure, TDR Probes component category brochure, and Time-Domain Reflectometry System brochure provide additional information about the TDR system components.
|Measurements Made||Volumetric water content (VWC) of porous media (such as soil), soil electrical conductivity (EC), rock mass deformation|
|Pulse Generator Output||250 mV into 50 Ω|
|Output Impedance||50 Ω ±1%|
|Time Response of Combined Pulse Generator & Sampling Circuit||≤ 85 ps|
|Pulse Generator Aberrations|
|Pulse Length||25.5 μs|
|Waveform Averaging||1 to 128|
|Operating Temperature Range||-40° to +85°C|
|Power Supply||Unregulated 12 Vdc (9.6 to 16 Vdc)|
150 mA maximum, USB powered (5 Vdc)
|Required Equipment||Measurement system|
|Dimensions||21.6 x 5.1 x 10.7 cm (8.5 x 2.0 x 4.2 in.)|
|Weight||0.79 kg (1.75 lb)|
|-NOTE-||20 to 10112 waveform values over chosen length|
Distance is Vp=1. Time is one-way travel.
Electrostatic Discharge Protection
|Air||±8 kV @ 2 Ω|
|Contact||±4 kV @ 2 Ω|
|Surge||±2 kV @ 2 Ω|
|During Measurement||120 mA|
|Sleep Mode||1 mA|